The Module 'Scope' is designed for implementing oscilloscopes by creating Device Classes.
- 1 Device Class development
- 1.1 Supported parameters
- 1.1.1 TriggerSource
- 1.1.2 TriggerCoupling
- 1.1.3 TriggerSlope
- 1.1.4 TriggerLevel
- 1.1.5 Acquisition
- 1.1.6 Average
- 1.1.7 SamplingRate
- 1.1.8 TimeRange
- 1.1.9 TimeRangeValue
- 1.1.10 TimeOffsetValue
- 1.1.11 Channel<i>
- 1.1.12 Channel<i>_Name
- 1.1.13 Channel<i>_Range
- 1.1.14 Channel<i>_Offset
- 1.1.15 Channel<i>_Impedance
- 1.1.16 Channel<i>_ACDC
- 1.1.17 Channel<i>_Coupling
- 1.1.18 Channel<i>_Probe
- 1.1 Supported parameters
Device Class development
An oscilloscope has many parameters that are needed before a certain configuration is reproduced. Sometimes it is easier to change parameters directly at the device. To not overwrite certain parameters when starting a measurment, we recommend to add "As is" to each ComboBox. In that case, the Device Class should not send any command to the oscilloscope.
A ComboBox to provide possible trigger sources such as the channels of the oscilloscope or an external trigger in port
A ComboBox to provide the trigger coupling that defines which mode is used to get triggered, e.g. it can be 'AC', 'DC', or high frequency ('HF').
A ComboBox to provide the trigger slope, e.g. a rising flank, a falling flank or both
A LineEdit to provide the trigger level at which the signal is recognized and the time is zeroed.
A ComboBox to provide how acquisition is done. Typically, the acquisition is done 'Continuous'. However, if the data related to a single trigger event needs to be measured, one should provide a mode like 'Single'.
A ComboBox to provide typical values that can be used for averaging, e.g. 1, 2, 4, 8 ...
A Line Edit to provide the numbers of sample per second.
A ComboBox to provide different modes that can be used to define the time span that is used for collecting the data.
A LineEdit to get the number that is specifying the time range. For example, if TimeRange is 'Time range [s]' one can insert the time span of the captured signal. If the TimeRange is 'Time scale [s/div]' one has to insert the time span of a single division.
A LineEdit to get the offset time in seconds to shift x-axis horizontally.
A CheckBox for each channel <i> that tells you whether the channel should be used or not.
The name that the user gave to channel <i>. Please use it do define self.variables accordingly.
The voltage range of channel <i>.
The voltage offset of channel <i>.
The impedance value of channel <i>.
The coupling of channel <i>. Please use Channel<i>_Coupling for version >= 1.5.4.x
The coupling of channel <i>, e.g. 'AC' or 'DC'
The probe of channel <i>. If not automatically detected one might have to provide options like '1:1' or '10:1'